|Place of Origin:||China|
|Minimum Order Quantity:||1|
|Packaging Details:||Plastic box|
|Delivery Time:||10-15 work days|
|Payment Terms:||L/C, T/T, Western Union, MoneyGram|
|Measuring Ranges:||10-60MPa||Impact Energy:||2.207J（0.225Kgf.m)|
|Dimensions:||Φ54X278mm||Standards:||ASTM C 805, BS 1881:202, Etc.|
Concrete Test Hammer CT-120 Concrete Testing Equipment Concrete Rebound Hammer
Concrete Test Hammer applied to inspect compressive strength for general building components, bridges and all kinds of concrete components (include in slabs, beams, columns, bridges for concrete) .Our concrete test hammer can meet the international standards:ASTM C 805, BS 1881:202, NF P18-417, DIN 1048 and UNI 9189.
Suitable for testing a wide variety of concrete, rock and bricks
Light, flexible and simple operation
Friction adjusted by the pointer
Adopt stretching techniques to let the button working well
Supplied with carborundum stone to prepare test surface
2. Technical Specification
Measuring strength ranges: 10-60MPa
Normal impact energy: 2.207J(0.225Kgf.m)
Strike hammer stroke: 75mm
The friction force of pointer slider: 0.5N±0.8N
Spherical radius of strike rod : 25mm±1mm
The rebound values calibrated on steel anvil: 80±2 Dimensions: Φ54*278mm
|Measuring strength ranges||10-60MPa|
|Normal impact energy||2.207J(0.225Kgf.m)|
|Strike hammer stroke||75mm|
|The friction force of pointer slider||0.5N±0.8N|
|Spherical radius of strike rod||25mm±1mm|
|The rebound values calibrated on steel anvil||80±2 Dimensions: Φ54*278mm|
3. Standard accessory
Concrete Test Hammer 1 pc(with portable wooden box)
Desiccant 1 bag
Millstone (with plastic box)
With the instrument's technical information
Product Certification of inspection
|Concrete Test Hammer||1PC|
|Millstone (with plastic box)||1|
|With the instrument's technical information||1|
|Product Certification of inspection||1|
The impact plunger (1) generates a recoil when it deploys. Always hold the concrete test hammer in both hands!
• Position the concrete test hammer perpendicular to the test surface.
• Deploy the impact plunger (1) by pushing the concrete test hammer towards the test surface until the pushbutton springs out.
Always hold the concrete test hammer in both hands, perpendicular to the test surface, before u trigger the impact!
Each test surface should be tested with at least 10 impacts. The individual impact points must be spaced at least 20 mm apart.
• Position the concrete test hammer perpendicular to and against the test surface. Push the concrete test hammer against the test surface at moderate speed until the impact is triggered (a high beep acknowledges registration).
• Repeat this procedure for the whole measurement series.